STEEN Solution™ Leaflet
For reliable objective ex vivo assessment of lungs
STEEN Solution™ is a buffered extracellular solution optimally designed to perfuse the isolated donor lung during ex vivo lung perfusion. STEEN Solution™ includes human serum albumin to provide an optimal colloid osmotic pressure and dextran 40 to coat and protect the endothelium from excessive leucocyte interaction. STEEN Solution™ is intended for assessment of isolated lungs after removal from the donor body for eventual transplantation into a recipient.
STEEN Solution™ is designed to facilitate prolonged evaluation and promote stability of isolated lungs ex vivo. The STEEN Solution™ perfusate and circuitry can maintain stable lung function, without edema formation, for up to 10 hours at 37 °C 7, 12, 18-22.
About four of every five lungs offered for transplantation are currently rejected by present selection criteria 1,2. Rejected lungs from marginal and extended donors have been successfully used for transplantation for almost two decades and the concept has now become routine practice3,4,5, 12, 16-18. The number of transplants of lungs perfused ex vivo is rapidly increasing with clinical EVLP programs being established world-wide.
With EVLP, the perfusion circiut mimics in-vivo conditions for the lung; the ventilated lung is perfused with deoxygenated STEEN Solution™, with or without red blood cells, and the critical parameters of gaseous exchange, pulmonary vascular resistance and other key variables under normothermic conditions are monitored. Variables such as pressures, flow rate, dynamic compliance, and small airway resistance are frequently used as markers of organ status.
The lungs are enclosed in the transparent container, XVIVO Organ Chamber™, to maintain humidity.
The pulmonary artery of the closed circuit is connected to a pump, oxygenator and heat so that the perfusate – STEEN Solution™ (with or without red blood cells) – assumes physiological temperature and partial gas pressures. A leucocyte filter is connected before the inflow to prevent leucocyte-induced tissue injury.
Since the lungs are generally cold before perfusion, initial flow rates must be very low, gradually increasing as the temperature increases. The flow should never exceed 4 l/min. Careful ventilation is begun as the temperature of the perfusate outflow reaches 32 °C.
Functional assessment can begin when the circuit has reached the steady state – (perfusion and ventilation parameters have reached the intended values and with the temperature of 37 °C,), monitoring arterial and venous blood gases, end-tidal carbon dioxide and a range of hemodynamic parameters.
To Order: REF 19004; 500 mL bottle
Content: 1 bottle contain 500 ml STEEN Solution™
Packaging: PETG bottle
Storage: Store dark at +2 to + 8oC in an upright position. Use only unopened and undamaged packages.
Shelf life: At least three months from date of shipment.
Intended Use: STEEN Solution™ is intended for assessment of isolated lungs after removal from the donor in preparation for eventual transplantation into a recipient.
Bacterial Endotoxins EU/ml <0.25
Osmolality (mOsm/kg) 275 – 315
pH 7.1 – 7.7
*In the US, the XPS™ with STEEN™ Solution is indicated for use in flushing and temporary continuous normothermic machine perfusion of initially unacceptable excised donor lungs during which time the ex vivo function of the lungs can be reassessed for transplantation.
The flexible comprehensive EVLP platform
Convenient, reliable, sterile accessories
Provides a method for evaluating lungs by EVLP
Single use, sterile disposable container
Single use, sterile disposable product